Megohmmeter or insulation resistance tester is a special type of ohmmeter used to measure the electrical resistance of insulators. Insulating components, for example cable jackets, must be tested for their insulation strength at the time of commissioning and as part of maintenance of high voltage electrical equipment and installations. Acceptable insulator resistance values are typically 1 to 10 megohms, depending on the standards referenced.
Resistance to be measured is connected across the terminals i. When the current is supplied to the coils then they have torque in opposite directions. If resistance to be measured is high, no current will flow through the deflecting coil, the controlling coil, will, therefore, set itself perpendicular to the magnetic axis and hence set the pointer to the infinity.
If the resistance to be measured is small, high current flow through the deflecting coil and the resulting torque sets the pointer to zero. For the intermediate value of resistance, depending upon the torque production, the pointer is set at a point between zero and infinity. The hand-driven generator is of the permanent magnet type and it is designed to generate from to volts.
Testing voltage is produced by the rotation of the crank in a hand-operated megger, or by the battery in the case of an electronic-type megger. For testing a range up to V, equipment requires V DC is sufficient. The current coil or deflecting coil is series-connected and allows the electric current to flow through the circuit being tested.
Control and deflecting coil has a current limiting resistor connected in series so as to protect the external circuit in case of damages caused due to very low resistance. Testing voltage is produced by electromagnetic induction in case of hand-operated megger and by a battery in case of electronic type megger.
The deflection of pointer increases with the increases in voltage in the external circuit and also decreases with the increase in current. That is, resultant torque is inversely related to current and directly related to voltage.
A Megger consists of an e. The scale of the voltmeter is calibrated in ohms kilo-ohms or megohms, as the case may be. In measurements, the e. From Wikipedia, the free encyclopedia. This article has multiple issues. Please help improve it or discuss these issues on the talk page. Learn how and when to remove these template messages. This article needs more article-links to integrate with Wikipedia. Improve this article by adding relevant links to the existing text! January Learn how and when to remove this template message.
This article does not cite any sources. Please help improve this article by adding citations to reliable sources. Unsourced material may be challenged and removed. Electrical and electronic measuring equipment. Arbitrary waveform generator Digital pattern generator Function generator Sweep generator Signal generator Video-signal generator.
Categories : Measuring instruments. Hidden categories: Articles with too few wikilinks from January All articles with too few wikilinks Articles covered by WikiProject Wikify from January All articles covered by WikiProject Wikify Articles lacking sources from September All articles lacking sources Articles with multiple maintenance issues.
Instruments for low resistance measurement are key to field and laboratory applications requiring high accuracy. The instruments can measure bonding joints, circuit breakers and contact or fuse resistance.
Filter By. Hide Filters. View as. Showing 1 - 25 of Show 5 10 15 20 Save item Compare. Add to Cart Add to Quote. Low Resistance Ohmmeter Learn More. Micro-Ohmmeter Learn More.
DC Resistance Meter. View details. Micro-ohmmeter A with Printer Learn More. High Current Mic 5AA,0. Narrow Results By Filter By. Shopping Options Price. Compare Products item s. Remove This Item Compare. Clear All. In stock. Item ships in about Ships in weeks.
Item ships in about Call for availability. Item ships in about 1 - 2 Weeks. Item discontinued. Item ships in about Ships in days.A milliohm meter is a handy benchtop tool for measuring small electrical resistance values.
I used careful design techniques to cancel many error sources rather than resort to costly components. The milliohm meter is useful for tasks such as measuring transformer and inductor winding resistance, ammeter current shunts, and resistance of PCB tracks. The finished milliohm meter. The milliohm meter calculates the value of the resistor under test Rx by measuring the voltage across it and the voltage across a series-connected, known reference resistor Rr when driven by a test current.
A technique called synchronous rectification also known as a lock-in amplifier is used to enhance accuracy. The direction of the test current is alternated and measurements of Vx and Vr are made synchronously with the change in direction of the test current.
Micro Ohmmeters & Milliohm Meter
As we will see, this cancels a number of error sources and is easy to implement on the Arduino. Synchronous rectification can be thought of as narrowband filter at the switching frequency, implemented using a mixer multiplier at the switching frequency followed by a low-pass filter at DC averaging.
Normally, the switching frequency would be high enough say, 1 kHz to allow AC-coupled, high-gain amplifiers to be used and to move the filter passband well away from induced Hz AC line voltages. In this implementation, the relatively slow ADC conversion speed prevents us from using a high switching frequency.
However, we retain many other benefits of synchronous rectification with regard to reducing measurement error and we gain accuracy improvement in other ways. An Arduino is used to control the synchronous rectification, read voltages Vx and Vr, and then compute and display the test resistor value.
The circuitry. The source and sink legs are alternately driven high and low to produce the test current. A look inside the meter. Series resistors and film capacitors on the analog inputs provide some overload protection and noise filtering. Each measurement consists of two cycles of synchronous rectification, with samples per cycle for a total of samples. An OLED module with an I2C interface is used for the display, although other options can be substituted with corresponding code changes.
The meter is powered by a 9-V battery. Battery voltage is read through one of the analog input ports. Measurements are initiated with the push of the test switch to maximize battery life and minimize self-heating errors in the reference resistor. Each measurement takes roughly 2 s. OLED for displaying data. The meter is built using purchased modules and a small piece of protoboard for the shield. The ADC and display modules are available from multiple sources, and you can use any Arduino module of your choosing.
The photos and layout are for the Pro Mini. Keep the ADC analog input wiring short and away from the processor.Request Free Catalog. The four-wire Kelvin bridge method eliminates test lead resistance to provide measurement accuracy up to 0. Our engineers regularly review and improve the features of our instruments so you can always expect AEMC Micro-Ohmmeters to be the highest quality. Utilizing a four-lead Kelvin method of testing, our instruments accurately measure very low resistances in the micro-ohm range within 0.
Our powerful DataView software, with customizable templates and professional data reports, is included to provide valuable insights and stores up to measurements model dependent saving time and money.
We created the following one-page comparison document to help you select the best Micro-Ohmmeter for your specific needs. Or e-mail your questions to our tech team. Our knowledgeable and friendly service team provides the best support in the industry. We seek to understand your inquiry or feedback in a respectful and responsive way.
We're happy to provide a demonstration with our technical experts. Contact us at or email us at sales aemc. Selectable test currents of 10A, 1A, mA, and 10mA, with a polarity reversal function. Tests both resistive and inductive material, and stores up to 99 measurements.
Three test modes are available; resistive instantaneous testinductive continuous testand auto repetitive test. Stores up to measurements. Features optimized filters and protection for measurements in electrical substations. Designed for measuring very low contact resistances of high voltage circuit breakers, switches, busbars, and other units. Join our Mailing List Email:. Product Catalog Request Free Catalog.
Key Features Measurement resolution down to 0. Data storage and report generation Comprehensive Data Analysis Software Our powerful DataView software, with customizable templates and professional data reports, is included to provide valuable insights and stores up to measurements model dependent saving time and money.
Micro-Ohmmeter Comparisons We created the following one-page comparison document to help you select the best Micro-Ohmmeter for your specific needs. Micro-Ohmmeter Model Cat. AEMC Instruments. All Rights Reserved. Customer Service Technical Support 1. Email Address.Discussion in 'Electronic Basics' started by Bart, Nov 23, Connect with us. Electronics Forums.
LCR meter Reply to Thread. Search Forums Recent Posts. Scroll to continue with content. Nov 23, 1. The motors are DC-servo with brushes, 3-phase AC servos voltand stepper motors.
The meter generates both 1khz and hertz and the display shows the henrie, millihenries or microhenries and also a "Q" value don't know what that is. Is there a way to figure out from a motor nameplate what my meter should read if the motor is good?
Megohmeters / Insulation Testers
AC motor's nameplates tell how many poles, voltage rating, amp rating, and horsepower. A DC motor's nameplate tells voltage, amp rating, and torque. The meter also can show resistance in ohms at either 1khz or hz. Thanks in advance, Bart. Nov 23, 2. Nov 24, 3. Nov 24, 4. Nov 24, 5.
Nov 24, 6. Nov 24, 7. Basically, "Q" is the ratio of inductive reactance to resistance, so an inductor generating ohms of inductive reactance and having a resistance of ohms has a Q of ten. Since inductive reactance is different at different frequencies and resistance is basically the same, Q generally increases with frequency up to a point.Milliohm and miniaturized scale ohm meters are accuracy test instruments equipped for estimating low resistance values in circuits, much of the time down to divisions of micro-ohms—less than one-millionth of an ohm.
These are the resistance meters to measure very low and very high resistance. Omni controls is a top supplier of measuring equipment including ohm meters, milliohm meters, digital micro-ohm meter from various leading brands such as Megger, Sonel Test and Measurement, Extech, Fluke, AEMC, Reed instruments and a lot others to offer you high precision, best protection and easy operability with the best next-generation test and measurement solutions.
Browse through our website to learn about digital micro-ohmmeter or analog ohm meters for sale at the best prices. Extech Battery Powered Milliohm Meter. Microohmmeter MMR Milliohm Meter Extech Heavy Duty Vibration Meter.
Milliohmmeter Battery Powered. Ohm Meter Low Resistance. Ohmmeter Low Resistance. Login Remember me. Lost your password? Showing 1—20 of 33 results Sort by popularity Sort by average rating Sort by latest Sort by price: low to high Sort by price: high to low.
Add to Wishlist. Product added! Browse Wishlist. The product is already in the wishlist! Quick View. Micro Ohmmeters Milliohm Meter Micro Ohmmeters Milliohmmeter Battery Powered.Understanding every aspect of the electricity network, from generation and transmission to distribution and consumption, is what it takes to be the world leader in cable test, diagnostic and fault location solutions.
Million Megohm Meters
Armed with this invaluable knowledge and supported by the extensive experience of our application engineers, we are able to tailor our services and product delivery to ensure our customers can always keep the power on! Read More. At the heart of Megger are its people. Find out more about us. Feel good testing.Fluke 1507 Motor Insulation and low ohm Testing
Discover a solution today. Megger test instruments. Learn more. The backbone of power distribution. Register Today and Stay Updated.
Cable testing Circuit breaker testing Electrician's testers Relay and protection testing Transformer testing Insulation testing Resistance, battery and PQ testing. Stories and updates. A kit for earth testing Electrical Tester Online.
We are all constantly adapting to the impacts related to COVID pandemic, particularly as the situation continues to Megger would like to inform our customers and business partners that areas of our business have the potential to be Megger Acquires Power Diagnostix News. Global electrical test equipment expert Megger has acquired Power Diagnostix of Aachen, Germany, a leader in the Since the invention of the transformer, the design, quality and efficiency of this magnificent machine has been improved Megger offers all-in-one testing for motors News.
Megger has combined all of the functions frequently required for testing electric motors in a unique and innovative Read more articles.
Technical articles Document. Visit technical library. PAT hand-held portable appliance tester animation. TDR introduction, function and operation. How to get the best from an MFT View all videos.